If you are starting to experience stalling issues with your Chemyx syringe pump system, there are a few questions to determine the root cause and get your motorized syringe pump working again.
Does the pump stall without any syringes?
If not, then the stall may have been caused by too much back pressure.
Try using a smaller syringe and/or setting a slower flow rate to minimize stall occurrences.
Is your pump 3+ years old?
If yes, then it might be due for a repair. You can email firstname.lastname@example.org with the SN of your pump. That number is on the back of the motorized syringe pump, below the barcode. We will send you the RA number and repair shipping instructions. The average turnaround time for repairs is one week.
Is your stall rod/lock nut too close to the end of the channel?
The stall rod/lock nut is to protect your syringe from over-plunging. You can adjust the rod by turning the knobs on the side of the pusher block. For the locknut, you can adjust it with the Allen wrench found in the back of your pump, above the power supply.